HP33120A MANUAL PDF
Service Guide. Publication Number (order as manual set) The security code is set to “HP” when the function generator is. Note: Unless otherwise indicated, this manual applies to all Serial Numbers. The Agilent .. For information on using the Phase-Lock Option for the HP A. Microsoft® Windows® included. Agilent A. 15 MHz Function /. Arbitrary Waveform Generator. Note: Unless otherwise indicated, this manual applies to all .
|Published (Last):||24 November 2005|
|PDF File Size:||1.48 Mb|
|ePub File Size:||19.34 Mb|
|Price:||Free* [*Free Regsitration Required]|
HP 33120A User Manual
Chapter 3 Features and Functions Output Configuration Hp333120a Configuration This section contains information to help you configure the function generator for outputting waveforms. Page 8 The front-panel display will light up while the function generator performs its power-on self-test. This information is manyal directly related to waveform generation but is an important part of operating the function generator.
Used only in the internal triggered burst mode. The default frequency is 1 kHz for all functions. For example, you can store such information as the last calibration date, the next calibration due date, the instrument serial number, or even the name and phone number of the person to contact for a new calibration. Freq 1 Enable the frequency modify mode. You are then exited from the menu. Page 42 Chapter 2 Front-Panel Menu Operation To trigger a burst or sweep To trigger a burst or sweep You can issue triggers from the front-panel for burst modulation and frequency sweeps using single trigger or internal trigger.
Certain combinations of burst count and carrier frequency are not allowed. Be sure to read the security code rules on page before attempting to secure the function generator.
Page 38 Chapter 2 Front-Panel Menu Operation To output a burst waveform To output a burst waveform You can configure the function generator to output a waveform with a specified number of cycles, called a burst.
You may never have to change some of the parameters discussed here, but they are provided to give you the flexibility you might need.
Chapter 3 Features and Functions Arbitrary Waveforms Arbitrary Waveforms There are five built-in arbitrary waveforms stored in non-volatile manuaal. Page Only the last six characters are recognized from the front panel, but all eight characters are required. The following steps show you how to change the amplitude to 50 mVrms. Chapter 3 Features and Mnual Calibration Overview To Secure Against Calibration You can secure the function generator against accidental calibration either from the front panel or over the remote interface.
Page 20 Front-Panel Menu Operation Chapter 3 Features and Functions Amplitude Modulation AM Modulating Waveform Frequency The function generator will accept an internal modulating signal, an external modulating signal, or both. Quick Start One of the first things you will want to do with your function generator is to become acquainted with its front panel. We have written the exercises in this chapter to prepare the function generator for use and help you get familiar with some of the front-panel operations.
The rate is set to baud when the function generator is shipped from the factory.
At this point, the pulse waveform is being output from volatile memory. The output amplitude is Output Amplitude automatically adjusted and no error is generated if you change the continued output termination. In this example, you will enable the function generator to automatically recall the power-down state when power is turned on.
Page IEEE interface and an interface. Notice that the annunciator flashes, indicating that the displayed percentage is the depth also called percent modulation. You can sweep up or down in frequency, and with either linear or logarithmic spacing. Page 19 Chapter 1 Quick Start To store the instrument state To verify that the state was stored properly, you can turn the power off before recalling the state.
HP A USER MANUAL Pdf Download.
To print the manual completely, please, download it. Page 26 Chapter 2 Front-Panel Menu Operation A front-panel menu tutorial Menu Example 1 The following steps show you how to turn on the menu, move up and down between levels, move across the choices on each level, and turn off the menu. SineSquare, Triangle, Ramp, manaul Arbitrary waveform.
Press or execute TRIG: This enables you to recall the entire instrument configuration with a single command from the remote interface or with just a few key presses from the front panel. You can set the offset to a positive or negative number with the restrictions shown below.
The function generator is shipped from the factory with a mAT fuse installed. Only one interface can be enabled at a time. Features and Functions You will find that this chapter makes it easy to look up all the details about a particular feature of the function generator.
Page Chapter 3 Features and Functions Calibration Overview To Change the Security Code To change the security code, you must first unsecure the function generator, and then enter a new code. Page 86 Chapter 3 Features and Functions Frequency Sweep Start Frequency and Stop Frequency The start frequency and stop frequency set the upper and lower frequency bounds manuao the sweep.
You can output the burst at ho33120a rate determined by the internal rate generator or the signal level on the rear-panel Ext Trig terminal. Page 72 Chapter 3 Features and Functions Burst Modulation Burst Modulation You can configure the function generator to output a waveform with a specified number of cycles, called a hp333120a. The function generator will accept only an internal modulating signal no external source is available.
Each waveform can contain between 8 and 16, data points.